IEC TR 60601-4-1:2017(E) is intended to help a manufacturer through the key decisions and steps to be taken to perform a detailed risk management and usability engineering processes for medical electrical equipment or a medical electrical system, hereafter referred to as MEE or MES, employing a degree of autonomy (DOA). This document provides a definition of DOA of MEE or MES and a medical robot, and also provides guidance on: - methodologies to perform the risk management process and usability engineering for an MEE or MES with a DOA; - considerations of basic safety and essential performance for an MEE and MES with a DOA; and - identifying the use of DOA, and similar concepts in existing ISO/IEC standards dealing with MEE or MES with the goal to facilitate alignment of standards by consistent use of the concept of DOA; and - distinguishing between medical robots, and other MEE and MES. Unless specified otherwise, this document considers MEE and MES together. The manufacturer of an MEE or MES with a DOA is expected to design and manufacture an MEE or MES that fulfils its intended use and does not have unacceptable risk throughout its life-cycle. This document provides guidance to help the manufacturer in complying with the requirements of IEC 60601-1:2005 and IEC 60601-1:2005/AMD1:2012 for MEE and MES with DOA. The document is also intended as guidance for future standard writers. There are no prerequisites to this document.
Status: PublishedPublication date: 2017-05
Edition: 1Number of pages: 80
Technical Committee: ISO/TC 299 Robotics
Buy this standard
|std 1 320|
|std 2 320||Paper|